Ebay IC Tester Instructions
The ubiquitous IC/Component tester found on Ebay usually comes without any instructions or documentation:
The unit I purchased came with a one page, 2-sided photocopy instruction sheet, so I thought I would share it here:
As far as I can tell, this unit will identify and test the following:
Common linear and logic IC’s
Some transistors; seems to be limited to bipolar
It does not appear to identify:
Three terminal regulators
Thanks for reading,
Tektronix CG5001/CG5011 Pulse Rise Time Measurement
I wanted to measure the rise time of the normal and fast edge pulses in a new acquired Tektronix CG5011 calibration generator. The equipment line is as follows:
Tektronix TM5006 mainframe
Tektronix CG5011 Calibration Generator
Tektronix CG5001 Calibration Generator
Tektronix/Tegam 015-0611-01 Pulse Head
HP 83480A Communications Analyzer
HP 83483A Electrical Sampling Module (20GHz)
Background: The original 015-0311-00/01 pulse head was used in combination with the CG551AP and CG5001 calibration generators. The 015-0611-00/01 pulse head is the later version used on the CG5010 and CG5011 calibration generators. The primary differences between the later 0611 and earlier 0311 are:
The 0611 has faster edge rise time of <150 ps or < 160 ps for the Tegam version
The 0311 has slower edge rise time of <200 ps
The 0611 is larger (taller) than the 0311
The 0611 made by Tegam has an SMA output connector
The 611 made by … Read More »
Tektronix 7904A S52 Pulse Response
This post documents my investigations into the following vintage electronic gear:
Tektronix 7904A Mainframe (500 MHz Bandwidth)
Tektronix S52 Pulse Head (tr<25ps)
Tektronix S4 Sampling Head (tr < 25ps)
Tektronix 7S12 TDR/Sampler Plug-in
Here is the test setup:
HP 8903B Audio Analyzer PCB Pad Replacement
Visual inspection of my recently acquired HP 8903B Audio Analyzer revealed that the polymer pads on the bottom side of the main PCB were seriously deteriorated:
Besides making an unsightly mess, this residue may form corrosive products which, over time, could damage the PCB. Also, the pads no longer perform their bottom cover to PCB isolation function. So all the old pads and residue had to be removed and new pads installed. Lintless wipes, isopropyl alcohol, and nitrile gloves were the chosen tools.
The residue cleaned up completely, and new pads were installed. I won’t need to be concerned with this failure mode again.
Thanks for reading.
Tektronix 7854 Waveform Keyboard Rebuild
From the Tek Wiki:
The Tektronix 7854 Waveform Processing Oscilloscope, introduced in 1980, is a 400 MHz combined analog / digital mainframe in the 7000 series that takes two vertical and two horizontal 7000-series plug-ins. In addition to a conventional analog (real time) scope, it contains an equivalent-time sampling 10-bit digitizer and a programmable waveform processor with GPIB interface. The waveform processor can be programmed using a detachable calculator-style keyboard, using postfix operators (“reverse Polish notation” like HP calculators) that operate on entire waveforms, not just individual numbers.
Tek 7854 Mainframe #2 after cleaning
It is to the repair of the detachable keyboard that this post is devoted. I purchased a keyboard for a reasonable price only to find that some keys were intermittent, and some did not work at all.
Opening up the keyboard showed that it was filled with debris:
The … Read More »
The uTracer Tube Tester and Curve Tracer
I have completed building my uTracer (micro Tracer) computer based vacuum tube tester and curve tracer. This unit, designed by Ronald Dekker described on his website:
The uTracer is much more than a simple tube tester; it is also a curve-tracer. It measures both the anode as well as the screen currents for a range of specified bias conditions, and arranges the measurements in meaningful sets of curves. From these curves parameters like the transconductance or the output resistance can then be extracted and plotted as a function of bias. Alternatively the uTracer can be used as a tester which extracts important parameters like Ia, Rp, gm and um in your favorite bias point in a matter of seconds. http://dos4ever.com/uTracer3/uTracer3_pag0.html
Ronald Dekker makes this tester available as a kit, with detailed assembly and testing instructions. One of … Read More »
Tektronix DSA602A Digitizing Signal Analyzer Part 1
I recently had the opportunity to purchase one of these units, and I am thoroughly enjoying the process of checking its operation and getting to know how to use it. This unit is a laboratory grade, very accurate digitizing oscilloscope from about 1990. From the Tek Wiki entry:
3 amplifier bays, up to 12 channels (using 11A34 plug-ins)
8 bits; averaging of N acquisitions increases by log2(N) up to 14 bits
DSA601, max 1 GS/s; DSA602, max 2 GS/s (one channel, left plugin)
512 to 32768 samples
50 ps/Div to 100 s/Div
7.2 bit up to 10 MHz, 6.3 bit @ 250 MHz, 5.7 bit @ 500 MHz, 4.5 bit @ 1 GHz
magnetic deflection color computer monitor, 10″ diagonal (130 mm × 168 mm), 552×704 pixels, 22×11 infrared touch-screen grid
90-132 V or 180-250 V, 48-72 Hz, 465 W (DSA601A) / 585 W (DSA602A)
Automated signal measurements (rise, fall, period, frequency, width, delay, phase, duty … Read More »
Tektronix Scope-Mobile Cart Repair Part 3
This is the conclusion to my repair of a Tektronix 202-2 Scope-Mobile Cart. Recall that the problem was three fractured trunnion mount bosses which rendered the cart useless (and unsafe) under load. My proposed solution was to place the bosses into compression by putting a 6mm countersunk bolt through the trunnion mount concentric with the existing hole in the boss. By using appropriate sized washers between the top of the boss and the frame member I ensure that the boss is placed into compression and the frame member is securely clamped. This avoids the tensile loading which fractured the bosses initially.
The holes in the trunnion mount were countersunk to fit the flat head socket bolts:
The cart was reassembled with the repaired trunnion mounts:
Felt lining was added to the plug in storage compartment:
Finally, the cart … Read More »
Tektronix 556 Dual Beam Oscilloscope Testing: Vertical Bandwidth
I have been doing some basic testing of my Tektronix 556 Dual Beam Oscilloscope, and I have documented some parts of this process. I have also captured images of the traces, which I will share here.
Test Setup: Tektronix 556, 1A2 plug-in, Timebase A, Tektronix SG 503 Generator, and Tektronix TG 501 Time Mark Generator
I found the -3db vertical bandwidth of the 1a2/556 to be 65 Mhz, which is to be expected for the 50 Mhz rated 1A2.
TM 500 Module with Plug ins
50 kHz Reference Signal 6v peak to peak
5 MHz Sine Wave 6v peak to peak
20 MHz Sine Wave 6v peak to peak
40 MHz Sine Wave 4.8v peak to peak
50 MHz Sine Wave 4.3v peak to peak
65 MHz Sine Wave 4.2v peak to peak
I will continue this testing by investigating the absolute maximum trigger … Read More »
Tektronix 556 Fan Motor Repair Part 2
In this continuation of Tektronix 556 Fan Motor Repair Part 1, I have removed the motor and mount assembly and disassembled it. Now I need to figure out the problem and rebuild the motor.
Tektronix 556 Fan Motor Overview, mounted in place
Tektronix 556 Fan Motor removed from mounts. The Tektronix part number is 147-0029-00.
Tektronix 556 Fan Motor removed from mounts.
Tektronix 556 Fan Motor removed from mounts.
The next few images show the disassembled parts of the motor:
Rotor, end bells and bushing assembly
End bell inside, retainer, spring washer felt and bushing.
End bell and bushing assembly
Rotor and spacers
At this point I found several issues by visual inspection:
The original rear bushing had been replaced, probably due to excessive internal wear.
The replacement itself had excessive internal wear and it did not fit in the retainer properly due to its shape.
The replacement … Read More »