Tag: Dual Beam Oscilloscope
Tektronix 556 Dual Beam Oscilloscope Testing: Vertical Bandwidth
I have been doing some basic testing of my Tektronix 556 Dual Beam Oscilloscope, and I have documented some parts of this process. I have also captured images of the traces, which I will share here.
Test Setup: Tektronix 556, 1A2 plug-in, Timebase A, Tektronix SG 503 Generator, and Tektronix TG 501 Time Mark Generator
I found the -3db vertical bandwidth of the 1a2/556 to be 65 Mhz, which is to be expected for the 50 Mhz rated 1A2.
TM 500 Module with Plug ins
50 kHz Reference Signal 6v peak to peak
5 MHz Sine Wave 6v peak to peak
20 MHz Sine Wave 6v peak to peak
Tektronix 556 Dual Beam Oscilloscope Images
What follows is a series of images of my Tektronix 556 Dual Beam Oscilloscope displaying various waveforms. I am learning about this scope and how to use it.
A brief description of the unit can be found here: Tektronix 556
Thanks for reading.
Tektronix 556 Dual Beam Oscilloscope
I had recently the unique opportunity to acquire a complete Tektronix 556 Oscilloscope setup. This comprised the oscilloscope itself, the matching scope cart, original manual, probes and 12 plug-in units. This in itself is rare enough, but in addition, this particular scope has the optically desirable blue phosphor CRT and Tektronix C-27 Camera Bezel. This 50 MHz bandwidth oscilloscope was the state of the art when it was introduced in the mid 1960’s: Tekwiki Tek 556
This will be the first of many posts and updates as I work my way through checking and restoring this beautiful scope.
And now some images: