In a previous post, I had documented the condition of several RIFA Y Safety capacitors, showing clearly the cracked condition of the outer case. This condition is a precursor to the well documented failure of this style of RIFA capacitor. Reportedly, these problems have been solved since Kemet squired RIFA in 2007. Of course the equipment which contains these potentially failing capacitors is typically older than 2007.
For example, I recently replaced six (two from each instrument) of these capacitors from the following equipment:
HP 3245A (1992)HP3245A (1992)HP 3458A (1989)
HP 3245A (1992) Original RIFA Safety Capacitors. The line filter is on the bottom right.
HP 3458A (1989) Original RIFA Safety Capacitors. The line filter cutout is on the bottom right.
Each and every RIFA Safety Capacitor I removed from this equipment showed cracks in the case.
Cracked RIFA … Read More »