Tag: Tektronix TG501
Tektronix 556 Dual Beam Oscilloscope Testing: Vertical Bandwidth
I have been doing some basic testing of my Tektronix 556 Dual Beam Oscilloscope, and I have documented some parts of this process. I have also captured images of the traces, which I will share here.
Test Setup: Tektronix 556, 1A2 plug-in, Timebase A, Tektronix SG 503 Generator, and Tektronix TG 501 Time Mark Generator
I found the -3db vertical bandwidth of the 1a2/556 to be 65 Mhz, which is to be expected for the 50 Mhz rated 1A2.
TM 500 Module with Plug ins
50 kHz Reference Signal 6v peak to peak
5 MHz Sine Wave 6v peak to peak
20 MHz Sine Wave 6v peak to peak